Please use this identifier to cite or link to this item:
|Title:||A simple laboratory set-up for the fringe-projection method|
|Source:||Muhd, I.R.B.Z.A., Jie, G.L.M., Wee, M., Lai, T., Fu, Y., Shang, H.M. (2001). A simple laboratory set-up for the fringe-projection method. Proceedings of SPIE - The International Society for Optical Engineering 4588 : 316-319. ScholarBank@NUS Repository. https://doi.org/10.1117/12.468718|
|Abstract:||A fringe-projection method for three-dimensional surface contouring using a commercial beam-splitting cube is described. A distinct advantage of this method over the Michelson interferometer lies in its ability to generate stable fringes with lax vibration isolation. Furthermore, the density of the fringes can be varied by changing the incidence angle of the laser beam. This enables the measurement of both large and small components.|
|Source Title:||Proceedings of SPIE - The International Society for Optical Engineering|
|Appears in Collections:||Staff Publications|
Show full item record
Files in This Item:
There are no files associated with this item.
checked on Apr 18, 2018
checked on Mar 12, 2018
Items in DSpace are protected by copyright, with all rights reserved, unless otherwise indicated.