Please use this identifier to cite or link to this item:
|Title:||Demonstration of Angle Widening Using EyeCam After Laser Peripheral Iridotomy in Eyes With Angle Closure|
|Citation:||Perera, S.A., Quek, D.T., Baskaran, M., Tun, T.A., Kumar, R.S., Friedman, D.S., Aung, T. (2010). Demonstration of Angle Widening Using EyeCam After Laser Peripheral Iridotomy in Eyes With Angle Closure. American Journal of Ophthalmology 149 (6) : 903-907. ScholarBank@NUS Repository. https://doi.org/10.1016/j.ajo.2010.01.013|
|Abstract:||Purpose: To evaluate EyeCam in detecting changes in angle configuration after laser peripheral iridotomy (LPI) in comparison to gonioscopy, the reference standard. Design: Prospective comparative study. Methods: Twenty-four subjects (24 eyes) with primary angle-closure glaucoma (PACG) were recruited. Gonioscopy and EyeCam (Clarity Medical Systems) imaging of all 4 angle quadrants were performed, before and 2 weeks after LPI. Images were graded according to angle structures visible by an observer masked to clinical data or the status of LPI, and were performed in a random order. Angle closure in a quadrant was defined as the inability to visualize the posterior trabecular meshwork. We determined the number of quadrants with closed angles and the mean number of clock hours of angle closure before and after LPI in comparison to gonioscopy. Results: Using EyeCam, all 24 eyes showed at least 1 quadrant of angle widening after LPI. The mean number of clock hours of angle closure decreased significantly, from 8.15 ± 3.47 clock hours before LPI to 1.75 ± 2.27 clock hours after LPI (P < .0001, Wilcoxon signed rank test). Overall, gonioscopy showed 1.0 ± 1.41 (95% CI, 0.43-1.57) quadrants opening from closed to open after LPI compared to 2.0 ± 1.28 (95% CI, 1.49-2.51, P = .009) quadrants with EyeCam. Intra-observer reproducibility of grading the extent of angle closure in clock hours in EyeCam images was moderate to good (intraclass correlation coefficient 0.831). Conclusions: EyeCam may be used to document changes in angle configuration after LPI in eyes with PACG. © 2010 Elsevier Inc. All rights reserved.|
|Source Title:||American Journal of Ophthalmology|
|Appears in Collections:||Staff Publications|
Show full item record
Files in This Item:
There are no files associated with this item.
checked on Sep 12, 2018
WEB OF SCIENCETM
checked on Sep 4, 2018
checked on Jun 15, 2018
Items in DSpace are protected by copyright, with all rights reserved, unless otherwise indicated.