Please use this identifier to cite or link to this item:
|Title:||The sampling limit in fluorescence microscopy||Authors:||Heintzmann, R.
Optical transfer function
|Issue Date:||Feb-2007||Citation:||Heintzmann, R., Sheppard, C.J.R. (2007-02). The sampling limit in fluorescence microscopy. Micron 38 (2) : 145-149. ScholarBank@NUS Repository. https://doi.org/10.1016/j.micron.2006.07.017||Abstract:||Sampling in fluorescence microscopy is treated using the concept of the three-dimensional (3D) optical transfer function (OTF). The border of the OTF frequency surface defines the required minimum sampling. The shape of the OTF is derived from simple considerations and valid for far-field high numerical aperture, vector theory. Optimal regular sampling is achieved by a hexagonal grid in 2D, and corresponding hexagonal structures, body-centered cubic (bcc) and hexagonal close-packed (hcp) structures, in 3D. As compared to standard (rectilinear grid) sampling a reduction of 13.4% in 2D and 29.3% in 3D can be achieved with optimized sampling. This reduction in data size is also accompanied by an imaging speed improvement, a reduction of sample bleaching, and can lead to imaging with better signal to noise ratio. © 2006 Elsevier Ltd. All rights reserved.||Source Title:||Micron||URI:||http://scholarbank.nus.edu.sg/handle/10635/67321||ISSN:||09684328||DOI:||10.1016/j.micron.2006.07.017|
|Appears in Collections:||Staff Publications|
Show full item record
Files in This Item:
There are no files associated with this item.
checked on Jul 29, 2020
WEB OF SCIENCETM
checked on Jul 22, 2020
checked on Aug 3, 2020
Items in DSpace are protected by copyright, with all rights reserved, unless otherwise indicated.