Please use this identifier to cite or link to this item:
|Title:||Enzyme-mediated amperometric biosensors prepared via successive surface-initiated atom-transfer radical polymerization|
|Citation:||Zhang, Z.B., Yuan, S.J., Zhu, X.L., Neoh, K.G., Kang, E.T. (2010-01-15). Enzyme-mediated amperometric biosensors prepared via successive surface-initiated atom-transfer radical polymerization. Biosensors and Bioelectronics 25 (5) : 1102-1108. ScholarBank@NUS Repository. https://doi.org/10.1016/j.bios.2009.09.023|
|Abstract:||The development of enzyme-mediated amperometric biosensors on the indium-tin oxide (ITO) glass electrode via surface-initiated atom-transfer radical polymerization (ATRP) was investigated. A trichlorosilane coupling agent, containing the sulfonyl halide ATRP initiator, was immobilized initially on the ITO electrode surface for consecutive surface-initiated ATRP of ferrocenylmethyl methacrylate (FMMA) and glycidyl methacrylate (GMA). Glucose oxidase (GOD) was subsequently immobilized on the modified ITO electrode surface via coupling reactions between the epoxide groups of GMA and the amine groups of GOD. The surface composition after each functionalization step was ascertained by X-ray photoelectron spectroscopy (XPS). With the introduction of redox-P(FMMA) block as the electron-transfer mediator, the enzyme-mediated ITO electrode exhibits high sensitivity, as revealed by cyclic voltammetry measurement. The sensitivities of the ITO-g-P(GMA-GOD)-b-P(FMMA) and ITO-g-P(FMMA)-b-P(GMA-GOD) electrodes are about 3.6 μA/(mM cm2) (in the linear concentration range 0-5 mM of glucose) and 10.9 μA/(mM cm2) (in the linear concentration range of 0-17 mM of glucose), respectively. For both biosensors, the steady-state response time and the detection limits are estimated to be less than 20 s and 0.4 ± 0.1 mM of glucose concentration, respectively. Furthermore, the spatial effect of the redox mediator on the electrode surface is revealed by the fact that the block copolymer brush-functionalized ITO electrode with P(FMMA) as the inner (first) block is more sensitive to glucose than that with P(GMA) as the inner block. © 2009 Elsevier B.V. All rights reserved.|
|Source Title:||Biosensors and Bioelectronics|
|Appears in Collections:||Staff Publications|
Show full item record
Files in This Item:
There are no files associated with this item.
checked on Jan 14, 2019
WEB OF SCIENCETM
checked on Jan 14, 2019
checked on Dec 21, 2018
Items in DSpace are protected by copyright, with all rights reserved, unless otherwise indicated.