Please use this identifier to cite or link to this item:
|Title:||Automated hollow fiber-protected dynamic liquid-phase microextraction of pesticides for gas chromatography-mass spectrometric analysis|
|Citation:||Hou, L., Shen, G., Lee, H.K. (2003-01-24). Automated hollow fiber-protected dynamic liquid-phase microextraction of pesticides for gas chromatography-mass spectrometric analysis. Journal of Chromatography A 985 (1-2) : 107-116. ScholarBank@NUS Repository. https://doi.org/10.1016/S0021-9673(02)01827-7|
|Abstract:||Dynamic liquid-phase microextraction (LPME) controlled by a programmable syringe pump was evaluated for extracting pesticides in water prior to GC-MS analysis. A conventional microsyringe with a 1.3-cm length of hollow fiber attached to its needle was connected to a syringe pump to perform the extraction. The microsyringe was used as both the microextraction device as well as the sample introduction device for GC-MS analysis. The attached hollow fiber served as the "holder" and "protector" of 3 μl of organic solvent. The solvent was repeatedly withdrawn into and discharged from the hollow fiber by the syringe pump. Pesticides were extracted from 4-ml water samples into the organic solvent impregnated in the hollow fiber. The effects of organic solvents, plunger movement pattern, agitation and extraction time were investigated. Good repeatabilities of extraction performance were obtained, with the RSD values ranging from 3.0% (alachlor) to 9.8% (4-chlorophenol) for the 14 pesticides; most RSD values were under 5.0%. The method provided a 490-fold preconcentration of the target pesticides. The limits of detection were in the range of 0.01-5.1 μg/l (S/N=3) in the GC-MS selected ion monitoring mode. In addition, sample clean-up was achieved during LPME because of the selectivity of the hollow fiber, which prevented undesirable large molecules from being extracted. A slurry sample (mixture of 40 mg soil/ml of water) containing seven pesticides was extracted using this method which also gave good linearity and precision (most RSDs<7.0%, n=3). © 2002 Elsevier Science B.V. All rights reserved.|
|Source Title:||Journal of Chromatography A|
|Appears in Collections:||Staff Publications|
Show full item record
Files in This Item:
There are no files associated with this item.
checked on Jan 12, 2019
WEB OF SCIENCETM
checked on Jan 2, 2019
checked on Nov 24, 2018
Items in DSpace are protected by copyright, with all rights reserved, unless otherwise indicated.