Please use this identifier to cite or link to this item:
|Title:||Predicting the life and design stresses of medical plastics under creep conditions|
|Source:||Teoh, S.H. (1994). Predicting the life and design stresses of medical plastics under creep conditions. ASTM Special Technical Publication (1173) : 77-86. ScholarBank@NUS Repository.|
|Abstract:||Polymers are finding increasing use in biomedical engineering applications. One of the major problems in designing with polymers is predicting when the material will fail and what stress values should be used in the design of components in order that the material will not fail prematurely. Recent work on modeling the creep rupture behavior of more than 15 polymers has produced a breakthrough in predicting the failure times as well as the upper stress and lower stress limits of the polymers. The upper stress relates to the maximum stress that can be applied to the material before instantaneous fracture. The lower stress limit relates to the minimum stress that can be sustained by the material without failure and, in some cases, this also corresponds to the fatigue endurance limit of the polymer. The creep rupture times of some medical plastics such as high density polyethylene (HDPE), polycarbonate (PC), polypropylene (PP), polyoxymethylene (POM), polyvinylchloride (PVC), polysulphone (PSU), and polymethylmethacrylate (PMMA) were analyzed at 20 °C in air. In order to optimize the parameters for creep rupture prediction, a nonlinear regression analysis program was used. Special attention was made for Delrin (POM from Du Pont) which is of interest in cardiovascular applications related to wear of mechanical heart valves. This material was tested in air at 37, 60, and 80 °C. The effect of saline solution was also studied. Here, the lower stress limit was reduced from 20 MPa to 5 MPa. All the specimens brittle fractured. This may account why Delrin used in the artificial hip joint prosthesis where the contact stresses exceed this limit was known to wear severely. However, in the disc of the mechanical heart valve no failure occurs because the stresses were probably below the lower stress limit.|
|Source Title:||ASTM Special Technical Publication|
|Appears in Collections:||Staff Publications|
Show full item record
Files in This Item:
There are no files associated with this item.
checked on Feb 16, 2018
Items in DSpace are protected by copyright, with all rights reserved, unless otherwise indicated.