Please use this identifier to cite or link to this item:
|Title:||Comparisons of the handheld autorefractor, table-mounted autorefractor, and subjective refraction in Singapore adults|
|Citation:||Farook, M., Venkatramani, J., Gazzard, G., Cheng, A., Tan, D., Seang-Mei, S. (2005-12). Comparisons of the handheld autorefractor, table-mounted autorefractor, and subjective refraction in Singapore adults. Optometry and Vision Science 82 (12) : 1066-1070. ScholarBank@NUS Repository. https://doi.org/10.1097/01.opx.0000192344.72997.7c|
|Abstract:||Purpose. The purpose of this article was to compare the Retinomax with the table-mounted autorefractor and subjective refraction in Singapore adults. Methods. Adults (n = 100) attending a tertiary eye hospital clinic were examined by an optometrist. First, subjective refraction testing was performed using a trial lens set, followed by handheld autorefractor tests using the Nikon Retinomax and the table-mounted autorefractor (Topcon RM8000B). Spherical equivalent and vector components of astigmatism were analyzed: J0 (Cartesian astigmatism) and J45 (oblique astigmatism). Results. The Retinomax autorefractor readings (mean = -4.69 D) were more minus compared with the table-mounted autorefractor (mean = -4.05 D) and subjective refraction (mean = -3.90 D). There were significant differences in J0 and J45 for comparisons between subjective refraction and Retinomax autorefraction, and table-mounted autorefraction and Retinomax autorefraction. Conclusion. The Retinomax autorefractor measures were more minus compared with the table-mounted autorefractor and subjective refraction. The Retinomax autorefractor is not recommended for research purposes, unless in remote inaccessible areas where a portable instrument is necessary and cycloplegia is not possible. Copyright © 2005 American Academy of Optometry.|
|Source Title:||Optometry and Vision Science|
|Appears in Collections:||Staff Publications|
Show full item record
Files in This Item:
There are no files associated with this item.
checked on Jun 18, 2018
WEB OF SCIENCETM
checked on Jun 5, 2018
checked on May 25, 2018
Items in DSpace are protected by copyright, with all rights reserved, unless otherwise indicated.