Please use this identifier to cite or link to this item:
|Title:||Comparison of eye-tracking success in laser in situ keratomileusis after flap creation with 2 femtosecond laser models|
|Authors:||Luengo Gimeno, F.|
|Citation:||Luengo Gimeno, F., Chan, C.M.L., Li, L., Tan, D.T.H., Mehta, J.S. (2011-03). Comparison of eye-tracking success in laser in situ keratomileusis after flap creation with 2 femtosecond laser models. Journal of Cataract and Refractive Surgery 37 (3) : 538-543. ScholarBank@NUS Repository. https://doi.org/10.1016/j.jcrs.2010.10.039|
|Abstract:||Purpose: To determine the efficiency of an eye tracker after laser in situ keratomileusis (LASIK) flap creation with 1 of 2 femtosecond laser models. Setting: Tertiary referral center, Singapore National Eye Center, Singapore. Design: Randomized clinical trial. Methods: The LASIK flap was created with an IntraLase (Group A) or a VisuMax (Group B) femtosecond laser. An Advanced Control Eye Tracker was initiated 3 times to obtain iris recognition. Eye tracking was considered successful if the eye movements could be followed despite the presence or absence of an opaque bubble layer (OBL). Univariate-multivariate logistic regression analysis was performed. Results: Preoperatively, the mean values (ranges) of the 87 eyes were sphere, -5.64 diopters (D) ± 2.17 (SD) (-9.75 to -0.25 D); cylinder 1.65 ± 1.63 D (-3.75 to 0.00 D); optical zone, 6.34 ± 0.20 mm (5.6 to 7.0 mm); keratometry, 43.48 ± 1.32 D (40.1 to 42.8 D); flap thickness, 109.30 + 5.21 μm (90 to 115 μm). Eye tracking was successful in 38 (90.5%) of 42 eyes in Group A and 43 (95.6%) of 45 eyes in Group B. No specific type of OBL was seen in either group. The LASIK was completed in all eyes. There was a statistically significant association between positive tracking and a smaller optical zone (P=.03). There were no statistically significant differences in eye tracking between the 2 femtosecond lasers. Conclusions: Eye tracking was achieved in more than 90% of cases after LASIK flap creation with 1 of 2 femtosecond laser models, even in the presence of an OBL. © 2011 ASCRS and ESCRS.|
|Source Title:||Journal of Cataract and Refractive Surgery|
|Appears in Collections:||Staff Publications|
Show full item record
Files in This Item:
There are no files associated with this item.
checked on Aug 11, 2018
WEB OF SCIENCETM
checked on Jul 25, 2018
checked on Jun 1, 2018
Items in DSpace are protected by copyright, with all rights reserved, unless otherwise indicated.